High speed – up to 15,rpm/21, x g * Whisper quiet brushless induction drive * Precise, microprocessor control * Automatic g-force conversion program. Hermle ZM-2 Microcentrifuge from Labnet International, Inc. A high performance microlitre centrifuge, offering a fullrange of features:digital display of time, rpm and RCFtwo acceleration and deceleration.
|Published (Last):||27 November 2004|
|PDF File Size:||18.56 Mb|
|ePub File Size:||4.74 Mb|
|Price:||Free* [*Free Regsitration Required]|
Never stay in the safety zone of 30 cm around the centrifuge or deposit dangerous materials inside this zone during centrifugation. Page 19 Error no.: Page 14 – Keyboard – Starting the centrifuge – “qu To print the manual completely, please, download it.
Hermle z m-2 Manuals
To open the centrifuge lid. Commercially available tubes or bottles made of glass or plastic. Width 28 cm Depth 37 cm Height The digital indications on the display will light up when the power is on.
Got it, continue to print. By pressing ‘print’ button you will print only current page. Power failure while rotor is in motion Action: The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations.
Refurbished Hermle Z M Centrifuge
Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Don’t have an account?
On the digital display “time min-sec” the letters “ro” rotor appears. Turn the speed knob 1see figure 4 till the required rotor code appears on the d i s p l a y. Operation Operation Operation Operation Operation 4. Should this happen, the centrifuge, rotor, buckets, adapters and the rotor chamber must be thoroughly cleaned and all broken particles removed immediately. Figure 8 If the green “fast” control lamp is lit, the unit accelerates and decelerates fast.
There is a distinction between two different kinds of errors. If the green “slow”control lamp is lit, the unit accelerates and decelerates very softly below 2 rpm.
Page of 20 Go.
G131038 Hermle Z 233 M-2 Labnet Centrifuge W/rotor
Enter text from picture: If this is not done, they could scratch the protective coating of the rotor. Preselection Of Operating Time 4.
Open and reclose the lid, restart the centrifuge Error herle Overcurrent or undervoltage due to power supply fluctuations Action The green control lamps 3 show which mode is activated. Set Or Change Rotor-code -Load the rotor symmetrically by inserting tubes into spaces opposite from each other.
The rotor must always be balanced in this fashion prior to operation.
Page 20 Error no.: With the centrifuge lid open you can preset the operating time with the knob 3. Don’t show me this message again.
Hermle LaborTechnik GmbH – Z 233 M-2 Microlitre Centrifuge
Description – The rotor is not installed securely on the motor shaft. The temperature rise depends on the rotor swing-out or angle rotor ,bucket type, ambient temperature, running time and the speed of the rotor.
During a run and with the centrifuge lid closed you have to press the “preset” key 1hold it and change operating time by turning the knob 3. General Information The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations.
Hremle only original spare parts!
Page 15 – Temperture Page 16 – Service and inspection of centrifuge Page 17 – Cleaning of centrifuge after breakage of Keyboard – Starting The Centrifuge – “quick”-key 4. Comments to this Manuals Your Name.
Service And Inspection Of Centrifuge 7. Table of Contents Add to my manuals Add.